Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.
Mychway Q Switched Nd Yag Laser Tattoo Eyebrow Removal With RED Target Light
- Q Switched Nd Yag Laser
- RED Target Light
ND: YAG Cosmetic System introduces gem-adjusting Q switch, which uses high energy by instant emission of beam to crush and remove the pigment effectively in the pathologically changed tissues. That is an explosion led by beam: the beam at the fixed wavelength enabled by the accumulated high energy can reach the pathologically changed tissues through epidermis in a very short time (only 6 ns) to instantly crush the respective color base, then the corresponding pigment group gets swollen and outburst at last due to high energy, and part of the pigment group springs out of the human body (at shallow epidermis), and the rest (at basis tissues) pigment group will become very small particle before swallowed by the human cell which will be finally discharged out of the human body through lymphatic system circulation. The pigment of the pathological tissues will fade in color till it vanishes. And the surrounding normal tissues will not be ruined since they will not absorb the at that fixed wavelength.
The system gem-adjusting Q switch can effectively cure the mixed pigments by pigment precipitation of extraneous source with pathological pigment change of endogenous source, inclusive of erroneous dark eyebrow-embroidery, lip-embroidery, lip-soak, eye line-embroidery, lip-line, tattoo (inclusive of tattoo by extraneous wounds), pigment precipitation by sensitive source, freckle on female's face (male's), coffee spot, old age spot, black mole, and birthmark in black, blue, purple, brown, red and etc. Opeartion course for different illness case as follows:
Opeartion course required ( cure times) Opeartion course interval Remark (requirements in surgery)
eyebrow-embroidery(eyeliner grain) 1 –2 times 30-60 days Eyebrow blow, eyeball protection
Eyebrow-embroidery ( eyeliner embroidery) 1-3 times 45-60 days Eyebrow blow, eyeball protection
Lip line (red dyed lip) 1-3 times 15-45 days Use 532 filter
Tattoo(caused by extraneous wounds) 1-4 times 30-60 days Blood infiltration will be fine on wounds
Freckle, age spot 1-4 times 15-30 days Blood infiltration will be fine on wounds
Birthmark, nevus/mole 3-7 times 45-90 days Must have Blood infiltration .
(1）Opeartion for tattoo in eyebrow
Customer could not open eyes during Opeartion , operator should adjust energy according to depth of tattoo color, hold the handle and move it to Opeartion area slowly to make customer feel acceptable in mentally. Right now, the speed of flash could be set 3HZ, keep head of the handle a two-finger distance to Opeartion area, start Opeartion from head of eyebrow to end of eyebrow. It is needed to treat again in 5-10 minutes if the color of tattoo re-appear.
(2）Remove eye line
The speed of flashes could be set 4-5HZ,adjust the energy of light to minimum energy, keep a one-finger distance between head of handle and Opeartion tment area. Operator should strictly protect customers’ eyes with panel for eyelid or medical pledget (panel for eyelid)of 1cm depth.
Opeartion should be started from down eye line, customer should deeply close eyes and keep eyeball up-sight inside. Operator should press the pledget (panel for eyelid) with forefinger and push up eyelid, peel down eyelid with middle finger to bare the eye line, treat it with flashes of 4-5Hz. Then, treat up eye line. Also customer should close eyes and keep eyeball down-sight inside. Operator should press the pledget (panel for eyelid) with middle finger and push eyelid down, push eyelid up with forefinger to bare the eye line, treat it with flashes of 4-5Hz.
(3）Remove lip line
The speed of flashes could be set 5Hz,keep a 3/4 finger distance between head of handle and Opeartion area. It is better to remove black, coffee, deep, red and brown lip line with 1064 head at first time and then, treat again with 532 head.
Notice: All tattoos in lip should be treated in 6 months after making. OR removal Opeartion will fail. It should have 45-60 days interval to continue next Opeartion until remove all.
The speed of flashes could be set 3Hz/5Hz.It is better to use high energy for Opeartion until Opeartion area seeps a little blood. It should have 45-90 days interval to continue next Opeartion until remove all.
(5）Remove spot and nevus
The speed of flashes could be set 3Hz/5Hz.Keep 3-4 fingers distance between head of handle and Opeartion area. It is better to use high energy for Opeartion until Opeartion area seeps a little blood. It is advised to use 1064 head for normal spot and nevus Opeartion , but for deep color spot and nevus, it is better to use 532 head for Opeartion . It should have a 15-30days interval to continue next Opeartion until remove all.
The speed of flashes could be set 3Hz/5Hz.It is better to use high energy for Opeartion until treatOpeartion ment area seeps a little blood. It is advised to use 1064 head for birthmark Opeartion , but for red and deep color birthmark, it is better to use 532 head for Opeartion . It should have 45-90 days interval to continue
10.4" color touch screen
Cooling condition: wind system+closed circulating+cooling system
type: Q-switch Nd: YAG
With RED target light to aim on the skin
The pulse energy: 0~1000mJ
Pulse width: 6ms
Laser wavelength: 1064nm / 532nm/BD-LS
Recurrence rate: 1, 2, 3, 4, 5, 6Hz
Power input: 110v~220v
1 × Main Unit
1 x Handpiece
2 x Tips (1064nm~532nm),
1 × Water funnel
1 × Package box
1 × Foot touch
1 × Protective goggle
1 × Power line (we stock 110V-240V power version and US.UK.UA.EU plugs for the country all of the world)
We provide the E manual. if you need read it first,Please contact us.
1.Innovative in technology,strict in technical detection,precise in manufacture,stable and effective in performance.
2.Special designed cooling system ensures constantly working.
3.Micro Computer counts the shoots.
4.Portable and aseismatic,convenient for transportation.
5.user friendly interface with 10.4" LCD display
6.No injury to normal tissue.
7.Remove pigmentation without side effects or scar.
8.Short Opeartion period with outstanding effect.